It can be used to promptly and accurately measure the luminous intensity and the illuminating position(or the optical axis deviation amount) of the high/low beam of vehicle headlight. With the lamp image alignment techique the single-machine high/low beam testing needs only 30-odd seconds.It is also used for the dual-lamp test at the testing station,the dual-machine test at the same time needs only 20-odd seconds.The testor has been equipped with a high performance emkedded industrial control computer for data collecting and analyzing;The touch screen and the supper bright LCD allows the operation,calibration and trouble-shooting to be more simple and convenient;It has a prompt function of adjusting lamp online so that the lamp-odjusting is more conveniently and promptly,It can be commected with the host computer via RS-232 and RJ45 port for convenient online testing.
|Measuring range||Luminous inlensity||0~120,000cd(a special measuring range can be tailored)|
|The optical axis deviation amount of light/low beam||up2°~down3°(up35cm/10m~down50cm/10m)|
|Indication error||Luminous intensity||±10%|
|Deviation error from the optical axis of high beam||±10′|
|Deviation error from the optical axis of low beam||±10′|
|Measuring distance||1 m|
|Power supply||AC (220V±22)V;(50±1)Hz|